It is expected that the Bidder will cooperate closely with PATH and target Hospitals for the new waste handling
building/area and will make available all needed technical information, drawings, etc.,
The following waste streams shall be treated in the new HCW treatment center(s):
Pharmaceutical waste including cytotoxic or genotoxic waste
Small amounts of chemical waste
TECHNICAL SPECIFICATION FOR PROCUREMENT OF “MEDICAL WASTE TREATMENT PLANT”
1.Hydroclave for Medical waste Treatment Plant: Quantity: 02 Nos.
The processor must be approved to sterilize & treat the following types of medical waste:
-Liquids: Blood, body fluids etc.
-Blood bags & tubing, gloves, Gowns, drapes, etc.
-Contaminated packaging: bandage etc.
-Sharps & Sharp containers
-Human and animal tissue (surgical and pathologic) –Based upon MWM regulations
-Cultures, stocks and glassware
The processor must comply with a documented efficacy level of 6log10 reduction in spore survival (99,9999%) of the
The processor should reduce the volume of the waste by 60% to 80% through complete dehydration and sterilization of
The processor must have ZERO harmful emission level and meet or exceed all standards required by the EU.
The whole waste treatment system should be offered as a complete unit consisting of following equipment and
(A)The sterilization processing unit
(D)Heavy duty external shredder-
(E)Conveyer belt-based on Manufacturer’s Recommendations
(A)Features and Specifications of the Sterilization Processor unit:
Technical specification of the processor:
Steam autoclave in combination with a heavy duty Fragmenting System inside the vessel/chamber
The vessel must be of Double Chamber
Capacity (kg/load) : minimum 200 kg/batch/cycle
Volume of 140-190 liters of bulk per cycle
Average Cycle time : 60 minutes
Operating temperature: 121◦C to 135◦C
Operating pressure : 1.1 Bar to 2.4 Bar